Resistance to Sliding in Orthodontics: A Systematic Review
Fax: +32 16 33 75 78; Email: firstname.lastname@example.org
The main concern of orthodontists has always been to achieve the most efficient, time-saving and successful treatment rendering optimal and satisfactory results for the patient. In order to achieve this target, it is important to fully understand the various factors and conditions involved in orthodontic tooth movement.The friction generated between bracket and archwire plays an important role in this respect. From a mechanical point of view, friction is defined as “the resistance to motion during sliding or rolling that is experienced when one solid body moves tangentially over another with which it is in contact” .In the last few decades, multiple orthodontic studies have reported on the role of friction in various clinical simulations. However, they actually reported resistance to sliding (RS), which is not interchangeable with the term “friction”. RS is a more comprehensive concept than friction because it also includes other components that resist sliding such as binding and notching . The term RS will be the only one used in this review except when static friction is expressed, which is the force required to initiate sliding or rolling. Many factors affect RS between bracket and archwire, such as the surface roughness of used materials  and bracket/archwire dimensions . It is assumed that RS may have an impact on some aspects regarding orthodontic treatment such as the rate of tooth movement  or increase in anchorage requirements. Nevertheless, there is a lack of conclusive clinical trials .
In Orthodontics, force systems are designed to produce tooth movement in two fundamental ways relative to the archwire in fixed appliances [4,6]. Firstly, tooth movements can be achieved by using archwires with loops and levers, where there is no actual sliding between bracket and archwire. Secondly, movements can result from a translation of tooth and bracket over a continuous archwire. The latter involves actual sliding between bracket and archwire, which results in frictional forces resisting the motion in an opposite direction (RS).
This review was conducted based on the second type of tooth movements, considering in vitro studies that evaluate the bracket/archwire system as it is the most commonly used fixed system to correct malocclusion in orthodontic practice. It is well known that although there are many clinical studies related to RS, none of them have been able to measure RS in vivo so far. Although in-vitro findings cannot be extrapolated to in-vivo settings due to the possible interaction of different biological variables, such as periodontal ligaments and bone condition, a considerable amount of very valuable information has been reported about RS measurement using different in-vitro trials. Inability to fully replicate the biological considerations possibly influencing RS is an obvious disadvantage of the invitro trials. However, a review and analysis of in-vitro studies may provide information on material and mechanical aspects involved in RS. To this end, we here reviewed in-vitro studies measuring RS.
The aim of this study was to systematically review in-vitro studies considering RS in orthodontics, to shed light on the role played by different mechanical and material-related aspects on RS.
Materials and Methods
The protocol for this review followed a review structure developed prior to the literature search. Reporting follows the PRISMA guidelines (www.prisma-statement.org).
Information sources and search strategy
An electronic search was performed in three different data bases, MedLine Database (Entrez PubMed, www.ncbi.nim.nih. gov), Web of Science (Inspec Database, apps.webofknowledge. com), and Grey Literature Report of the New York Academy of Medicine (www.greylit.org) (Table 1). The search included all articles published until the 26th of March 2015. A search string of keywords was developed based on the terms “friction”, “orthodontics” and “resistance”, which are the most commonly used keywords in the studies of interest (Table 1).
Table 1. Electronic search string used in this review Eligibility
Primary selection of studies was based on their title and abstract. Selected studies fulfilled the inclusion/exclusion criteria listed in Table 2.
The studies obtained with the established search string were screened and selected based on the title and the abstract. A single examiner performed the selection according to inclusion/exclusion criteria (Table 2). An additional manual search was performed using the reference lists of the included studies. Additionally selected studies fulfilled the inclusion/ exclusion criteria listed in Table 2.
A final selection was performed after a full text reading of the included studies. The previously used criteria (Table 2) were re-examined in the full text. Studies which did not aim to state a direct and measurable relation between RS and other related parameters were additionally excluded.
Table 2. List of inclusion and exclusion criteria Risk of bias in individual studiesRisk of bias in individual studies
The selected studies were subjected to an established quality assessment protocol (Table 3) and given a score. The scoring was done as follows: Good = 1, Average = 0.5 and Poor = 0. The scores of the 12 questions are shown in Appendix I. Accordingly, studies were grouped into three categories:
• Score of 5.5 or less: Low-quality papers
• Score 6 – 7.5: Low to moderate-quality papers
• Score of 8 or more: Moderate to high-quality papers
Data collection process
Figure 1 show the data extracted from the included studies. A data extraction sheet was developed and piloted. To facilitate comparison, studies were grouped according to the different testing situations used in the literature (Figure 2). These testing situations represented different factors or aspects that could affect or could be affected by RS during orthodontic treatment. Then, studies were compared per group descriptively. The first author collected the data and reviewed the studies collected. The second and corresponding author performed additional reviewing and cross-checking of findings.
Study selection and study characteristics
An overview of the electronic search and selection is shown in a flow diagram following the PRISMA guidelines (www. prisma-statement.org) (Figure 3). The initial search retrieved 1466 records from all data bases (1351 PubMed, 115 Inspec, 0 Grey Literature Report). Once duplicates were removed, 1420 records were screened. Additionally, 9 records were added after manual search. After applying the selection criteria basedFigure 1. Set of data extraction.on title and abstract (Table 2), 1039 records were excluded. Of the 390 full-text articles, 145 were excluded for not matchingthe purpose of the review. A total of 245 eligible studies were included in the present review.
Risk of bias of studies
The details of the quality assessment protocol are listed in Appendix I. Of the included studies, 17.55% were considered of a moderate to high quality, 25.71% were of low quality and 56.73% were of medium quality.Figure 2. Classification protocol of included studies according to parameters related to the RS.Figure 3. Illustrative PRISMA flow diagram revealing the general screening overviewTable 3. Quality assessment protocolSelected studies
The selected studies were found to relate many parameters to RS. All the selected studies investigated RS of differentbracket/archwire combinations in vitro. Related parameters varied in their impact on RS. A more comprehensive overview is reported in the discussion section.
In this review, a custom-made scale was used to assess the quality of the studies included (Appendix I). The studies werenumerous and very diverse in designs and protocols, and it was impossible to use one of the reported quality assessmentscales for this purpose. In addition, existing scales are not validated for the type of material researches in the present review. In this section, papers of higher quality and more related information were mainly discussed.The relation between RS and related material and mechanical parameters is summarized as follows:1. Bracket:
It is common in the literature to evaluate the properties of different bracket materials, dimensions or designs in relation to RS.
1.1. Bracket materials:
Brackets of different materials have different frictional properties due to differences in their chemical and morphological structure. It was widely reported that among different materials used for bracket fabrication, stainless steel (SS) brackets showed the least resistance to sliding due to smooth polished surfaces facilitating sliding [7-13]. On the other hand, aesthetic all-ceramic brackets were shown to have higher RS due to their porous, rough surfaces [8-10,12,14- 18]. Even within the same bracket material itself, RS differed according to manufacturing technique. It was reported that sintered SS brackets were associated with less RS than cast ones, again due to smoother surfaces . This may be an explanation of a proportional relation between RS and surface roughness of brackets. In terms of surface roughness of aesthetic brackets, glass-fiber reinforced brackets were reported to have less RS than ceramic reinforced ones . Similarly, the monocrystalline sapphire (MCS) and single crystalline alumina (SCA) brackets were reported to have less -or at least comparable- RS than the polycrystalline alumina brackets (PCA) due to smoother surfaces of MCS [8,11,14]. On the other hand, it was reported that monocrystalline ceramic brackets were associated with higher RS than polycrystalline ceramic brackets in a dry setting, although they have smoother surfaces [20-22]. Different findings suggested that surface roughness of bracket materials may well not be the only important aspect playing a role in the RS. Findings differed when other parameters prevailed related to RS such as angulation [14,22], wire surface conditions  or ligation force . Similarly, titanium brackets showed less RS thanstainless steel ones although they had a rougher surface. Differences in findings were reported to be due to differences in manufacturing process resulting in differences in surface chemistry . Additionally, the presence of an oxide film on the metal slot of composite brackets played a role in decreasing their static friction to levels comparable to stainless steel brackets as it acts as a barrier between the bracket slot and the metallic wire [20,26]. Furthermore, the sharp asperities of composite brackets led to less contact area between the bracket and the wire, lowering the force required to initiate sliding .
In addition, bracket materials of different hardness showed different RS. The harder SS brackets showed less RS compared to the softer titanium ones, which showed higher abrasion sensitivity . On comparing three different models of SSbrackets [SR-50 A, MiniDiamond® and Archist®], the SR-50A in the as-received state showed the lowest hardness values andthe highest RS .
1.2. Bracket design:
A new trend in designing brackets is the insertion of dissimilar materials into brackets, e.g., slots and caps of self-ligatingbrackets (SLBs) [9,29]. The aesthetic ceramic brackets are highly demanded for their superior esthetic properties. However, their rough surface and decreased hardness compared to the conventional SS brackets has led to the introduction of metal or silica slots into conventional designs. It was found that the introduction of metal or silica slots led to a marked reduction in RS due to their smoother surfaces than those of the all-ceramic brackets. Similar findings were reported when the metal insert is wide enough to prevent contact between test wire and slot walls. This was supported by consistent agreement in the literature [9,11,12,14-17,20,26,30].
When the caps of SLBs were made of dissimilar materials than the bracket material itself, RS behaved differently. SPEED® SLBshowed higher RS than other SS SLBs  because of its NiTi clip due to differences in coefficient of friction (COF) betweenNiTi and SS, which led to higher RS of SPEED® when there was no clearance between wire and bracket . Shape andmechanism of the cap of SLBs played a role in terms of RS. SLBs with a bell-shaped cap as Time 3® were reported to have higher RS than other SLBs with a little cap due to higher ligation force . SmartClip® SLB showed higher RS than other types of passive SLBs as it has two clips causing binding of the wire . However, those differences could be eliminated whenround wires were used instead of rectangular large ones . When SLBs with sliding caps (e.g. Damon®) were ligated to rectangular large wires, they created more contact with the wire leading to higher RS than other SLBs with caps sliding along a rail (e.g. Vision®) . In the first order of angulation, it was found that the passive SLBs showed a higher static frictionthan active SLBs. This was due to the non-flexible slides of passive SLBs, which did not extend against the wire pressure as did the flexible springs of the active SLBs (acted as stress absorbent).
Among the conventional ligated brackets, the design of the Tip-Edge® brackets resulted in less RS than the Mini-Diamond®brackets as the Tip-Edge® design allowed more wire clearance in the slot. This was consistent with research on Tip-Edge®and Transmission Straight Archwire® brackets , which showed that high critical contact angles (θc)a due to the edgeoffstructure led to less influence of the binding component (BI) on RS by expanding the passive configuration.b Anotherexample is the Synergy® brackets, which have a unique designallowing different manners of ligation. It was reported thatligating the inner wings of the Synergy® brackets leads to marked reduction in RS.*
Addition of bumps and rounded slot walls did not affect RS below a critical contact angle (θc) during tipping since RS isindependent of contact area. However, rounded slot walls increased θc leading to less impact of BI on RS but at theexpense of some control of root positions [37-40]. Rounded slots create more clearance between wire and bracket and allow offsetting the significant contribution of elastomeric ligatures to overall friction in the system . On the other hand, addition of bumps led to a higher BI component than conventional brackets due to bending of the wire in the slot before negotiating the slot walls .
Slot inclination may promote sliding depending upon the wire shape. Brackets with small deviation in the slot generated more contact with the rectangular archwires, leading to higher static friction . On the other hand, when those brackets were coupled to round archwires, they caused wires to slide easier, leading to less static friction .
1.3. Bracket width and slot depth:
Bracket designs of different dimensions also have an impact on RS. Wider brackets are associated with smaller interbracketdistances (IBDs) during the levelling stage of treatment, making the wire stiffer and hence decreasing θc, which leads to a higher BI component [33,41]. During labiolingual displacement, deeper slots allow more clearance between wire and bracket, and hence less RS . On the other hand, no correlation was reported between RS and slot depth or bracket width during horizontal displacement .Differences in findings may have been due to differences in the study design.
It was reported that the narrower brackets were associated with less RS as they offered more clearance for the wire duringtipping . A proportional relation between RS and bracket width can be concluded [43,44]. However, inconsistent findingswere reported because the ligation forces prevailed over other parameters related to RS. Some studies reported that wider brackets caused less RS than narrower ones by allowing less angulation change of the archwire [6,45,46].
1.4. Bracket slot size:
Some studies evaluated RS among brackets of different slot sizes and there is a consistent agreement that the slot size does not affect RS per se. It is generally believed that slot size depends more upon the bracket/wire combination [19,45,47-50]. It is
reported that the slot-filling bracket/archwire combinations produced the highest RS due to lack of clearance between wire and bracket . Wires placed in the slot using the central positioning (CP) method and compared with those visually positioned, were found to offer more clearance than the wires that were visually placed . These findings confirm that RS is mainly bracket/wire combination dependent, and that the slot size is not a determining variable per se.
1.5. Inter-bracket distance:
Another variant related to brackets that influences RS is the inter-bracket distance (IBDc). If the degree of malocclusionincreases, the IBDs decrease and the angulation between archwire and brackets increases, leading to stiffer wire and higher RS [31,47]. Hence, there is an inverse relation between IBD and RS consistent with some studies [43,51]. However, only one study reported that a change in IBDs to the extent generally encountered in canine retraction did not influence the force required to initiate sliding .
2.1. Wire material (surface conditions):
Differences in archwire surface conditions were shown to have an impact on RS depending on the material used. Severalstudies reported that SS wire showed the lowest RS followed by NiTi and βTi due to differences in their surface roughness[10,11,15,19,25,42,52-54]. Others also reported that βTi wires showed higher RS than SS ones, but comparable to or lowerthan NiTi ones [9,23]. Differences in findings could be due todifferences in medium and setup of testing. Additionally, βTi wires showed highest RS among NiTi and SS wires due to adherence to the slot surface, but SS wires showed slightly higher RS than NiTi wires suggesting that it could be due to differences in wires stiffness [17,55]. In addition, cobalt chromium (CoCr) and SS wires were reported to show less RS than NiTi and βTi wires due to differences in surface roughness .
Coating and ion implantation of wires for aesthetic reasons also had an impact on RS. It was reported that surface refinement by ion implantation or coatings of wires resulted in less RS than uncoated, untreated wires [40,56-58]. The newly introduced TMATM wires such as Honeydew and ionimplanted TMATM wires, showed lower RS than conventional TMATM wires although still higher than SS wires . Some studies also reported that TMATM wires showed higher RS than SS wires [14,40,60,61]. TMATM wires were reported to have high RS because of their titanium-rich layer, which breaks down, reacts, adheres, and breaks away resulting in stick-slip phenomena .
Coated wires were also reported to increase RS [20,62,63]. This is due to stripping or scratching of the coating during sliding [20,63] or due to adhesion that arises from molecular forces between the coating and the bracket surfaces .
2.2. Wire material (elastic properties):
Differences in mechanical or elastic properties of archwires have an impact on RS. A slight increase in RS with stiff archwires was reported during sliding with a cycle of relative tipping and up righting between bracket and archwire . This is consistent with some studies reporting that stiffer wires were associated with higher RS values if there was tipping between brackets and wires [8,9,64,65]. A clinical benefit of using flexible wires is that during alignment stage, misaligned or maloccluded teeth can easily negotiate greater angulations, resulting in lower RS values .
2.3. Wire hardness:
Hardness of wires differs and affects RS behavior of wires. A decrease in hardness values of wires is generally associated with an increase in abrasive wear. Accumulation of wear debris in turn leads to an increase of RS values [67-72].
2.4. Wire dimensions and shape:
It is widely accepted that wires of larger sizes are associated with higher RS rather than smaller ones as clearance between bracket and wire decreases [6, 8,10,11,16,17,19,23,29,34,42,43,50,52,53,55,57,60,64,66,73- 77]. RS depends mainly on the vertical dimension of the wire . As long as the vertical dimensions of the wires are smaller than the bracket slot size, and in the absence of angulation, the RS value is independent of the wire size [50,78]. However, smaller wires may still act unpredictly depending on the wire itself. Smaller βTi wires showed higher RS than larger ones due to tilting of wires under ligation force, resulting in a prominent edge and increased wear .
Considering different wire cross-section shapes, when clearance no longer exists between bracket and wire, rectangular wires showed higher RS than round wires did [19,27,29,79-81].
On the other hand, some authors reported that a difference in wire cross-section shapes had no influence on static friction[27,82]. In the absence of angulation or tipping, rectangular wires showed less clearance in the slot buccolingually than round wires, resulting in a higher ligation force withrectangular wires and, hence, higher RS value [11,40,83,84]. One study reported that a round wire of size 0.016-inch showed a virtually similar RS value of a rectangular wire of size 0.016×0.022-inch of the same material . Thus, for mesiodistal tooth movement, rectangular wires are preferred because of their additional feature of buccolingual root control.
3. Displacement in 3D: Translation and rotation:
3.1. Labio-lingual displacement:
Generally, an increase in labiolingual displacement results in a reduction of clearance between bracket and archwire, leading to a higher RS value. Static friction rises with an increase in labiolingual displacement especially when θc is exceeded[33,36,74,85]. Similarly, RS increases with augmented displacement. In other words, the bending moment produced by a wire and the resulting shear force during sliding predictably increase as the deflection increases above a certain critical angle θc. This varies depending on the bracket/wire combination used [74,86,87]. Only one study compared static friction between wire ligated on a convex arch and another ligated on a flat plate . The brackets set on a convex arch had a different buccolingual relation than those set on a flat plate without angulations. Furthermore, the wire ligated to a convex arch model showed a higher static friction than that ligated on a flat model because the angulation of the wire to the bracket increased the contact areas on friction components.
3.2. Occlusogingival displacement:
Some experimental set-ups evaluated RS under occlusogingival displacement, i.e. during the leveling stage of clinicaltreatment. Similar to other types of displacement, an increase in occlusogingival displacement led to a decreased clearance between bracket and wire, leading to a higher RS [16,33,65,74,89,90]. In such types of malocclusion, the common practice is to use more flexible archwires as they show a smaller BI component than stiff wires, as indicated above. RS was reported to increase with occlusogingival displacement but interestingly, both tested wires (L&H Titan® and Sentalloy®; super elastic NiTi) showed lower RS value at a 4 mm displacement than at a 3 mm one . It was suggested that the formation of stress-induced martensite resulted in lower stiffness and thus reduced BI component.
Several studies evaluated RS during second-order angulation or tipping. Many authors reported that RS markedly rises withincreasing tipping as clearance between wire and bracket no longer exists (above θc) and a larger contact area and frictioncomponent are created [7,8,14,22,34,35,63,64,66,75,81,91- 101]. The friction force (FR) seems to equal RS in the passiveconfiguration below θc,  while whenever clearance disappears and angulation exceeds θc, the BI component (moment) is additionally involved in RS . The BI moment depends on the mechanical properties or stiffness of the archwire as explained above. Wider periodontal ligament spaces cause the bracket tipping to increase, leading to higher RS value .
Some study designs simulated the biological retarding forces which face the root by the surrounding tissues during sliding. This retarding resistance acts in an opposite direction to the traction force, causing the tooth to rotate or to tip around itsbuccolingual axis . Virtually, that results in bracket tipping. An increase in the retarding forces was associated with arise in RS due to augmentation of load at contact points at a bracket/wire/ligature couple, i.e., increased tipping . This emphasizes the need for a passive levelling arch before applying a mesiodistal traction force, consistent with other studies that simulated retarding forces [45,46,99,102].
When a torque acts, the wire is twisted, leading to an increase of RS [103-105]. Also, RS rises with higher degrees of malocclusion, which is correlative with an increase in angulations in all planes of movement [31,47]. An increase of the inclination in the 3rd order is correlated with an increase in static friction when inclination rises above the critical angle[33,106]. However, the increase in static friction under torque is less than the one caused by tipping .
4.1. Ligature Materials:
Since ligatures are considered as a cornerstone in orthodontic treatment, the effect of different ligature materials on RS needs to be evaluated. It is widely reported that SS ligatures are associated with lower RS values than conventional elastomeric ligatures (CEL) [12,13,43,107-111]. This can be due to differences in surface properties between SS and elastomers. Also, the metal-to-metal contact offered by SS ligatures leads to smoother sliding due to high polishing of SS. However, SS ligatures may show RS comparable with CEL due to difficulties in standarization of SS ligation . When perturbations (gentle finger strokes on the wire simulating the occlusal displacing forces) were applied to the wire which was ligated by either SS ligatures or CEL, reduction in RS due to perturbations was independent of the change in ligature material .
Teflon-coated SS ligatures showed lower static friction than polyurethane elasomeric ligatures since the Teflon coat acts as a solid lubricant [8,13,114]. Similarly, silicon-lubricated elastomeric ligatures caused a lower static friction than the non-lubricated ones . Improvements in surface conditions of ligatures by coatings or lubrications resulted in lower RS values as noticed with slick-coated elastomeric ligatures (SuperSlick®) exhibiting lower static friction than other conventional elastomeric modules, as well as a better abrasion resistance, which influences the initiation of sliding . In agreement with this, some studies reported that slick-coated ligatures showed lower RS value than uncoated ligatures [117,118]. However, SS and Teflon-coated SS ligatures were reported to have superior sliding behavior compared to CEL and SuperSlick® ligatures [13,111].
4.2. Ligation Force (Ligation Technique):
Ligation force differs according to the ligation technique or the type of ligation. Several studies reported a direct relationbetween ligation force and RS, i.e., the higher the applied ligation force, the higher the RS value [11,24,83,84,119-121].
SLBs are assumed to produce a lower ligation force leading to a lower RS. Small round wires for both active and passive SLBs showed non-significant differences, while with larger rectangular wires the active SLBs showed higher RS than the passive ones. The active SLBs were affected by the increase in the buccolingual dimension of wires leading to higher RS . However, both SLBs show lower RS value than conventionally ligated brackets (CLBs) due to a lower ligation force [9,29,53,83]. As indicated above, other parameters such as bracket material [16,22,55], force decay of elastomeric ligatures [40,92] or bracket dimensions and design  may take the upper hand over ligation force, leading to inconsistent results. When tipping is involved, an increased RS value of SLBs is due to the BI component, and irrespective of the ligation force. However, active SLBs also showed higher RS during tipping due to the added higher FR value than passive SLBs (RS = FR + BI) [64,95].
In general, several studies reported that active SLBs showed a higher RS than passive SLBs [33,34,54,66,74,75,77,83,106,116,122]. Moreover, several studies reported that SLBs showed a lower RS than CLBs especially with small round wires [16,18,31,34,35,38,54,57,60,66,74,75,77,81,83,84,97,106,116].The low RS values obtained when ligating small round wires against SLBs suggest a good option for the alignment phaseduring orthodontic treatment. In contrast, it was also reported that SLBs showed higher RS than Mini-Twin® CLB .
Elastomeric ligatures have various designs and patterns of ligation influencing the ligation force. It was reported that theintroduction of a 45° bend into the elastomeric module reduced RS to comparable levels with the SS ligatures . Figureof-8 ligation method showed higher RS than the conventional ligation [13,109,114]. New designs of elastomeric ligatures, non-conventional elastomers, are considered as passive ligatures as they create less contact with wires. They showed lower ligation force and RS value than conventional designs [89,125,126]. Slide®, low-friction design of elastomeric ligatures, showed lower RS than the conventional one but only with round wires . The size of ligatures was shown not to affect RS significantly [82,121,127]. The Delta Force® CoCr bracket has multiple tie wings to allow different forms of ligation: minimum, medium and maximum ligation . Adirect rise in RS was reported at increasing ligation pattern.
5. Oral atmosphere and clinical use:
For many years a controversy has raged over whether saliva or fluid medium is a lubricant or an adhesive when a bracket is slid through an archwire. Several studies in the literature shed light on the hypothesized lubrication effect of saliva. A reduction of RS at 0° tipping was reported, while at 10°, lubrication was material-dependent . Hence, the lubrication effect of artificial saliva exists when clearance is present between brackets and wires. When clearance disappears, the sliding behavior depends on the chemical and surface properties of tested materials. This was explained by the squeezing out of saliva between bracket and wire when the angulation increases, leading to a direct contact between the studied couple [7,34].
On the other hand, some studies reported an overall increase in RS in the wet state, suggesting an adhesive effect played bysaliva against some bracket/wire couples [20,21,95,101,128]. This was explained by the nature of saliva, which is a polarliquid increasing the atomic attraction between ionic species of bracket and wire . Although they had the roughest surface among NiTi and SS wires, βTi wires showed the lowest increase in RS under wet conditions. This was explained by the fact that saliva fills up the irregularities in the βTi wire surface and makes it smoother (lubrication effect) . Similar findings were noticed with respect to SS and βTi wires . The sliding behavior is independent of the degree of saliva viscosity .
Prolonged exposure to an oral medium (saliva, heat, and humidity) may lead to an elastic degradation and an elastic force decay of elastomeric ligatures leading to lower RS value . It may also lead to changes in the structure and surfacecharacteristics of ligatures, affecting RS . However, polyurethane ligature material and slick coated ligatures werereported to be less prone to oral environmental changes than conventional latex [82,132] and uncoated ligatures .Similarly, SS ligatures are least affected by the oral medium .If the oral medium temperature in the case of austenitic NiTi wire increases, RS rises as well . This was explained by an increase in NiTi wire stiffness with higher temperatures leading to a rise in the BI component of RS .5.3. Wear and reconditioning:
Many studies reported that an increase in surface roughness and abrasion of test wires were associated with an oral useleading to an increase in RS [56,60,133-137].
In a similar context, the bracket of highest corrosion resistance (SR-50A) showed decreasing RS with increasing exposure tothe oral environment, while brackets with less corrosion resistance (MiniDiamond® and Archist®) showed increasing RS . This was explained by the augmented brackets surface corrosion products, which substantially affect RS. A higher increase in RS was noticed after immersion in artificial saliva at 6.75 pH or in a 0.2% acidified phosphate fluoride solution (APF) reflecting the severity of corrosion caused by APF. On the other hand, when 0.2% chlorhexidine solution was compared with artificial saliva, a non-significant difference between the two solutions was noticed . It was concluded that chlorhexidine-containing mouthwashes can be safely prescribed as a non-destructive prophylactic agent for orthodontic patients. Coatings on orthodontic wires may improve the corrosion resistance and, hence may have a beneficial effect on sliding behavior [70,140].
Wires repeatedly used showed higher RS due to wearing and the abrasion process, leading to a rougher surface. Wear debris generally only has a small influence on static friction, except at a high ligation force, where the static friction markedly increases because of wire bending .
A process of reconditioning of used archwires has a nonsignificant influence on static friction . On the other hand, reconditiong increases RS . On the contrary, it was reported that cleaning and conditioning of clinically used archwires by ultrasound or steel wool sponge resulted in elimination of accumulated debris and improvement of the surface properties, leading to reduction in RS . Air borne polishing caused a rougher, more irregular surface of polished brackets and, hence, increased RS [143,144].
5.4. Occlusal forces:
Few studies focused on the effects of occlusal displacing forces occurring during mastication on RS. However, a generalreduction in RS associated with occlusal displacing forces was reported [90,113,145-149]. This was explained as the forces causing the wire to unlock momentarily and to slide. However, that was inconsistent with tests using a loop and a bracket in the buccal sulcus to evaluate RS when patients were masticating a softened chewing gum. It was reported that mastication did not significantly reduce RS . This was possibly due to the position of the test bracket in the buccal sulcus, i.e., not in alignment with the occlusal plane.
As far as could be determined, none of the studies included in this review investigated whether plaque or pellicle affect RS or not.
5.6. Sliding Velocity:
Only two studies evaluated the influence of the sliding velocity on RS. RS values of SS and NiTi wires were reported to beindependent of sliding velocity, while CoCr wire showed a higher RS at lower velocities, and the opposite was noticed on βTi wire . An increase in RS with decreasing sliding velocity was reported . It was stated that Coulomb’s law of friction, i.e. “Friction is independent of sliding velocity”, was apparently not applicable at extremely low sliding velocities.
5.7. Bracket protectors:
Only one study investigated the difference in RS when two types of bracket protectors were used . The acetate protectorshowed a non-significant difference with non-protector state; while the temporary resin protector showed a slight increase in RS, possibly due to a larger contact area with the wire.
9. Loftus BP, Artun J, Nicholls JI, Alonzo TA, Stoner JA. Evaluation of friction during sliding tooth movement in various bracket-arch wire combinations. Am J Orthod Dentofac Orthop. 1999, 116 (3): 336-345.
13. Pattan SK, Peddu R, Bandaru SK, Lanka D, Mallavarapu K et al. Efficacy of Super Slick Elastomeric Modules in reducing Friction during Sliding: A Comparative in vitro Study. J Contemp Dent Pract. 2014, 15 (5): 543-551.
19. Vaughan JL, Duncanson MG, Jr., Nanda RS, Currier GF. Relative kinetic frictional forces between sintered stainless steel brackets and orthodontic wires. Am J Orthod Dentofac Orthop. 1995, 107 (1): 20-27.
20. Sadique S, Ramakrishna S, Batchelor AW, Bing CH. In vitro frictional behavior and wear patterns between contemporary and aesthetic composite orthodontic brackets and archwires. Wear. 2006, 261 (10): 1121-1139.
21. Pimentel RF, de Oliveira RS, Chaves M, Elias CN, Gravina MA. Evaluation of the friction force generated by monocristalyne and policristalyne ceramic brackets in sliding mechanics. Dental Press J Orthod. 2013,18 (1): 121-127.
27. Michelberger DJ, Eadie RL, Faulkner MG, Glover KE, Prasad NG, Major PW. The friction and wear patterns of orthodontic brackets and archwires in the dry state. Am J Orthod Dentofac Orthop. 2000, 118 (6): 662-674.
33. Yeh CL, Kusnoto B, Viana G, Evans CA, Drummond JL. In-vitro evaluation of frictional resistance between brackets with passive-ligation designs. Am J Orthod Dentofac Orthop. 2007, 131 (6): 704.e11-22.
34. Henao SP, Kusy RP. Evaluation of the frictional resistance of conventional and self-ligating bracket designs using standardized archwires and dental typodonts. Angle Orthod. 2004, 74 (2): 202-211.
36. Crincoli V, Perillo L, Di Bisceglie MB, Balsamo A, Serpico V, Chiatante F et al. Friction forces during sliding of various brackets for malaligned teeth: an in vitro study. ScientificWorldJournal. 2013, 2013: 871423.
41. Nucera R, Lo Giudice A, Matarese G, Artemisia A, Bramanti E et al. Analysis of the characteristics of slot design affecting resistance to sliding during active archwire configurations. Prog Orthod. 2013, 14:35.
42. Kapila S, Angolkar PV, Duncanson MG, Jr., Nanda RS. Evaluation of friction between edgewise stainless steel brackets and orthodontic wires of four alloys. Am J Orthod Dentofac Orthop 1990, 98 (2): 117-126.
49. Kusy RP, Whitley JQ. Resistance to sliding of orthodontic appliances in the dry and wet states: influence of archwire alloy, interbracket distance, and bracket engagement. J Biomed Mater Res. 2000, 52 (4): 797-811.
50. Willems G, Clocheret K, Celis JP, Verbeke G, Chatzicharalampous E et al. Frictional behavior of stainless steel bracket-wire combinations subjected to small oscillating displacements. Am J Orthod Dentofac Orthop. 2001,120 (4): 371-377.
53. Thomas S, Sherriff M, Birnie D. A comparative in vitro study of the frictional characteristics of two types of self-ligating brackets and two types of pre-adjusted edgewise brackets tied with elastomeric ligatures. Eur J Orthod. 1998,20 (5): 589-596.
54. Krishnan M, Kalathil S, Abraham KM. Comparative evaluation of frictional forces in active and passive self-ligating brackets with various archwire alloys. Am J Orthod Dentofac Orthop. 2009,136 (5): 675-682.
55. Cacciafesta V, Sfondrini MF, Ricciardi A, Scribante A, Klersy C et al. Evaluation of friction of stainless steel and esthetic self-ligating brackets in various bracket-archwire combinations. Am J Orthod Dentofac Orthop. 2003,124 (4): 395-402.
59. Cash A, Curtis R, Garrigia-Majo D, McDonald F. A comparative study of the static and kinetic frictional resistance of titanium molybdenum alloy archwires in stainless steel brackets. Eur J Orthod. 2004, 26 (1): 105-111.
61. Juvvadi SR, Kailasam V, Padmanabhan S, Chitharanjan AB. Physical, mechanical, and flexural properties of 3 orthodontic wires: an in-vitro study. Am J Orthod Dentofac Orthop. 2010,138 (5): 623-630.
62. Bandeira AM, dos Santos MP, Pulitini G, Elias CN, da Costa MF. Influence of thermal or chemical degradation on the frictional force of an experimental coated NiTi wire. Angle Orthod. 2011, 81 (3): 484-489.
63. Dickson JA, Jones SP, Davies EH. A comparison of the frictional characteristics of five initial alignment wires and stainless steel brackets at three bracket to wire angulations–an in vitro study. Br J Orthod. 1994, 21 (1): 15-22.
64. Thorstenson GA, Kusy RP. Effect of archwire size and material on the resistance to sliding of self-ligating brackets with second-order angulation in the dry state. Am J Orthod Dentofac Orthop. 2002,122 (3): 295-305.
65. Liaw YC, Su YY, Lai YL, Lee SY. Stiffness and frictional resistance of a superelastic nickel-titanium orthodontic wire with low-stress hysteresis. Am J Orthod Dentofac Orthop. 2007,131 (5): 578.e12-8.
68. Zufall SW, Kennedy KC, Kusy RP. Frictional characteristics of composite orthodontic archwires against stainless steel and ceramic brackets in the passive and active configurations. J Mater Sci Mater Med. 1998, 9 (11): 611-620.
69. Alfonso MV, Espinar E, Llamas JM, Ruperez E, Manero JM, Barrera JM et al. Friction coefficients and wear rates of different orthodontic archwires in artificial saliva. J Mater Sci Mater Med. 2013, 24 (5): 1327-1332.
72. Grosgogeat B, Jablonska E. Tribological response of sterilized and un-sterilized orthodontic wires. Materials Science & Engineering C, Biomimetic and Supramolecular Systems. 2006, 26 (2-3): 267-272.
74. Kim TK, Kim KD, Baek SH. Comparison of frictional forces during the initial leveling stage in various combinations of self-ligating brackets and archwires with a custom-designed typodont system. Am J Orthod Dentofac Orthop. 2008, 133 (2):187.e15-24.
77. Sims AP, Waters NE, Birnie DJ, Pethybridge RJ. A comparison of the forces required to produce tooth movement in vitro using two self-ligating brackets and a pre-adjusted bracket employing two types of ligation. Eur J Orthod. 1993, 15 (5): 377-385.
79. Ogata RH, Nanda RS, Duncanson MG, Jr., Sinha PK, Currier GF. Frictional resistances in stainless steel bracket-wire combinations with effects of vertical deflections. Am J Orthod Dentofac Orthop. 1996, 109 (5): 535-542.
81. Lee SM, Hwang CJ. A comparative study of frictional force in self-ligating brackets according to the bracket-archwire angulation, bracket material, and wire type. Korean J Orthod. 2015, 45 (1):13-19.
83. Tecco S, Di Iorio D, Cordasco G, Verrocchi I, Festa F. An in vitro investigation of the influence of self-ligating brackets, low friction ligatures, and archwire on frictional resistance. Eur J Orthod. 2007, 29 (4): 390-397.
86. Reznikov N, Har-Zion G, Barkana I, Abed Y, Redlich M. Measurement of friction forces between stainless steel wires and “reduced-friction” self-ligating brackets. Am J Orthod Dentofac Orthop. 2010, 138 (3): 330-338.
90. Seo YJ, Lim BS, Park YG, Yang IH, Ahn SJ et al. Effect of tooth displacement and vibration on frictional force and stick-slip phenomenon in conventional brackets: a preliminary in vitro mechanical analysis. Eur J Orthod. 2014.
95. Thorstenson GA, Kusy RP. Comparison of resistance to sliding between different self-ligating brackets with second-order angulation in the dry and saliva states. Am J Orthod Dentofac Orthop. 2002, 121 (5): 472-482.
97. Cordasco G, Lo Giudice A, Militi A, Nucera R, Triolo G et al. In vitro evaluation of resistance to sliding in self-ligating and conventional bracket systems during dental alignment. Korean J Orthod. 2012, 42 (4): 218-224.
103. Sims AP, Waters NE, Birnie DJ. A comparison of the forces required to produce tooth movement ex vivo through three types of pre-adjusted brackets when subjected to determined tip or torque values. Br J Orthod. 1994, 21 (4): 367-373.
112. Cordasco G, Farronato G, Festa F, Nucera R, Parazzoli E et al. In vitro evaluation of the frictional forces between brackets and archwire with three passive self-ligating brackets. Eur J Orthod. 2009, 31 (6): 643-646.
118. Leander D, Kumar JK. Comparative evaluation of frictional characteristics of coated low friction ligatures – Super Slick Ties with conventional uncoated ligatures. Indian J Dent Res. 2011, 22 (1): 90-94.
121. Tecco S, Tete S, Festa F. Friction between archwires of different sizes, cross-section and alloy and brackets ligated with low-friction or conventional ligatures. Angle Orthod. 2009, 79 (1):111-116.
122. Oliver CL, Daskalogiannakis J, Tompson BD. Archwire depth is a significant parameter in the frictional resistance of active and interactive, but not passive, self-ligating brackets. Angle Orthod. 2011, 81 (6): 1036-1044.
125. Franchi L, Baccetti T, Camporesi M, Barbato E. Forces released during sliding mechanics with passive self-ligating brackets or nonconventional elastomeric ligatures. Am J Orthod Dentofac Orthop. 2008, 133 (1): 87-90.
126. Galvao MB, Camporesi M, Tortamano A, Dominguez GC, Defraia E. Frictional resistance in monocrystalline ceramic brackets with conventional and nonconventional elastomeric ligatures. Prog Orthod. 2013,14: 9.
127. Ioi H, Yanase Y, Uehara M, Hara A, Nakata S et al. Frictional resistance in plastic preadjusted brackets ligated with low-friction and conventional elastomeric ligatures. J Orthod. 2009, 36 (1):17-22.
136. Normando D, de Araujo AM, Marques Ida S, Barroso Tavares Dias CG, Miguel JA. Archwire cleaning after intraoral ageing: the effects on debris, roughness, and friction. Eur J Orthod. 2013, 35 (2): 223-229.
138. Kao CT, Ding SJ, Wang CK, He H, Chou MY et al. Comparison of frictional resistance after immersion of metal brackets and orthodontic wires in a fluoride-containing prophylactic agent. Am J Orthod Dentofac Orthop. 2006, 130 (5): 568.e1-9.
139. Hosseinzadeh Nik T, Hooshmand T, Farazdaghi H, Mehrabi A, Razavi ES. Effect of chlorhexidine-containing prophylactic agent on the surface characterization and frictional resistance between orthodontic brackets and archwires: an in vitro study. Prog Orthod. 2013,14:48.
144. Parmagnani EA, Basting RT. Effect of sodium bicarbonate air abrasive polishing on attrition and surface micromorphology of ceramic and stainless steel brackets. Angle Orthod. 2012, 82 (2): 351-362.
149. Seo YJ, Lim BS, Park YG, Yang IH, Ahn SJ et al. Effect of self-ligating bracket type and vibration on frictional force and stick-slip phenomenon in diverse tooth displacement conditions: an in vitro mechanical analysis. Eur J Orthod. 2014, 37(5): 474-480.
150. Yanase Y, Ioi H, Nishioka M, Takahashi I. Effects of sliding velocity on friction: an in vitro study at extremely low sliding velocity approximating orthodontic tooth movement. Angle Orthod. 2014, 84 (3): 451-458.
151. de Lima Mendonca S, Praxedes Neto OJ, de Oliveira PT, dos Santos PB et al. Comparison of friction produced by two types of orthodontic bracket protectors. Dental Press J Orthod. 2014, 19 (1): 86-91.
153. Redlich M, Gorodnev A, Feldman Y, Kaplan-Ashiri I, Tenne R. Friction reduction and wear resistance of electro-co-deposited inorganic fullerene-like WS 2 coating for improved stainless steel orthodontic wires. J Mater Res. 2008, 23 (11): 2909-2915.
156. Katz A, Redlich M, Rapoport L, Wagner HD, Tenne R. Self-lubricating coatings containing fullerene-like WS 2 nanoparticles for orthodontic wires and other possible medical applications. Tribology Letters. 2006, 21 (2):135-139.
161. Khalid SA, Kumar V, Jayaram P. The comparison of frictional resistance in titanium, self-ligating stainless steel, and stainless steel brackets using stainless steel and TMA archwires: An in vitro study. J Pharm Bioallied Sci. 2012, 4 (Suppl 2): S203-11.
162. Kusy RP, Whitley JQ. Frictional resistances of metal-lined ceramic brackets versus conventional stainless steel brackets and development of 3-D friction maps. Angle Orthod. 2001, 71 (5): 364-374.
165. Thorstenson GA, Kusy RP. Effects of ligation type and method on the resistance to sliding of novel orthodontic brackets with second-order angulation in the dry and wet states. Angle Orthod. 2003, 73 (4): 418-430.
168. Huang SY, Huang JJ, Kang T, Diao DF, Duan YZ. Coating NiTi archwires with diamond-like carbon films: reducing fluoride-induced corrosion and improving frictional properties. J Mater Sci Mater Med. 2013, 24 (10): 2287-2292.
172. Suwa N, Watari F, Yamagata S, Iida J, Kobayashi M. Static-dynamic friction transition of FRP esthetic orthodontic wires on various brackets by suspension-type friction test. J Biomed Mater Res B. Appl Biomater. 2003, 67 (2): 765-771.
175. Thorstenson G, Kusy R. Influence of stainless steel inserts on the resistance to sliding of esthetic brackets with second-order angulation in the dry and wet states. Angle Orthod. 2003, 73 (2): 167-175.
176. Whitley JQ, Kusy RP. Resistance to sliding of titanium brackets tested against stainless steel and beta-titanium archwires with second-order angulation in the dry and wet states. Am J Orthod Dentofac Orthop. 2007, 131 (3): 400-411.
182. Meier MJ, Bourauel C, Roehlike J, Reimann S, Keilig L et al. Friction behavior and other material properties of nickel-titanium and titanium-molybdenum archwires following electrochemical surface refinement. J Orofac Orthop. 2014, 75 (4): 308-318.
188. Vinay K, Venkatesh MJ, Nayak RS, Pasha A, Rajesh M et al. A comparative study to evaluate the effects of ligation methods on friction in sliding mechanics using 0.022″ slot brackets in dry state: An In-vitro study. J Int Oral Health. 2014, 6 (2): 76-83.
193. Kusy RP, Tobin EJ, Whitley JQ, Sioshansi P. Frictional coefficients of ion-implanted alumina against ion-implanted beta-titanium in the low load, low velocity, single pass regime. Dent Mater. 1992, 8 (3):167-172.
196. Redlich M, Katz A, Rapoport L, Wagner HD, Feldman Y et al. Improved orthodontic stainless steel wires coated with inorganic fullerene-like nanoparticles of WS(2) impregnated in electroless nickel-phosphorous film. Dent Mater. 2008, 24 (12): 1640-1646.
197. Regis S, Jr., Soares P, Camargo ES, Guariza Filho O, Tanaka O et al. Biodegradation of orthodontic metallic brackets and associated implications for friction. Am J Orthod Dentofac Orthop. 2011,140 (4): 501-509.
203. Kahlon S, Rinchuse D, Robison JM, Close JM. In-vitro evaluation of frictional resistance with 5 ligation methods and Gianelly-type working wires. Am J Orthod Dentofac Orthop. 2010,138 (1): 67-71.
210. Stefanos S, Secchi AG, Coby G, Tanna N, Mante FK. Friction between various self-ligating brackets and archwire couples during sliding mechanics. Am J Orthod Dentofac Orthop. 2010,138 (4): 463-467.
211. Thorstenson GA, Kusy RP. Resistance to sliding of self-ligating brackets versus conventional stainless steel twin brackets with second-order angulation in the dry and wet (saliva) states. Am J Orthod Dentofac Orthop. 2001,120 (4): 361-370.
213. Wong JK, Romanyk DL, Toogood RW, Heo G, Carey JP et al. The effect of perturbations on resistance to sliding in second-order moments comparing two different bracket types. J Dent Biomech. 2014, 5:1758736014557500.
214. Yanase Y, Ioi H, Uehara M, Hara A, Nakata S et al. Comparison of the kinetic frictional force between conventional plastic brackets with thermoplastic low-friction module ligation and self-ligating brackets. World J Orthod.2009, 10 (3): 220-223.
216. Al-Khatib S, Berradja A, Celis JP, Willems G. In vitro friction of stainless steel arch wire-bracket combinations in air and different aqueous solutions. Orthod Craniofac Res. 2005, 8 (2): 96-105.
218. Chng CK, Foong K, Gandedkar NH, Chan YH et al. A new esthetic fiber-reinforced polymer composite resin archwire: a comparative atomic force microscope (AFM) and field-emission scanning electron microscope (FESEM) study. Prog Orthod. 2014, 15 (1): 39.
220. Guerrero AP, Guariza Filho O, Tanaka O, Camargo ES, Vieira S. Evaluation of frictional forces between ceramic brackets and archwires of different alloys compared with metal brackets. Braz Oral Res. 2010, 24 (1): 40-45.
222. Jakob SR, Matheus D, Jimenez-Pellegrin MC, Turssi CP, Amaral FL. Comparative study of friction between metallic and conventional interactive self-ligating brackets in different alignment conditions. Dental Press J Orthod. 2014, 19 (3): 82-89.
223. Kumar S, Singh S, Hamsa PRR, Ahmed S, Prasanthma et al. Evaluation of friction in orthodontics using various brackets and archwire combinations-an in vitro study. J Clin Diagn Res. 2014, 8 (5): Zc33-6.
224. Matarese G, Nucera R, Militi A, Mazza M, Portelli M et al. Evaluation of frictional forces during dental alignment: an experimental model with 3 nonleveled brackets. Am J Orthod Dentofac Orthop. 2008, 133 (5): 708-715.
225. Mendes Bde A, Neto Ferreira RA, Pithon MM, Horta MC, Oliveira DD. Physical and chemical properties of orthodontic brackets after 12 and 24 months: in situ study. J Appl Oral Sci. 2014, 22 (3): 194-203.
232. Cacciafesta V, Sfondrini MF, Scribante A, Klersy C, Auricchio F. Evaluation of friction of conventional and metal-insert ceramic brackets in various bracket-archwire combinations. Am J Orthod Dentofac Orthop. 2003,124 (4): 403-409.
234. Espinar E, Llamas JM, Michiardi A, Ginebra MP, Gil FJ. Reduction of Ni release and improvement of the friction behaviour of NiTi orthodontic archwires by oxidation treatments. J Mater Sci Mater Med. 2011, 22 (5): 1119-1125.
239. Kumar BS, Miryala S, Kumar KK, Shameem K, Regalla RR. Comparative Evaluation of Friction Resistance of Titanium, Stainless Steel, Ceramic and Ceramic with Metal Insert Brackets with Varying Dimensions of Stainless Steel Wire: An In vitro Multi-center Study. J Int Oral Health. 2014, 6 (5): 66-71.
240. Leite VV, Lopes MB, Gonini Junior A, Almeida MR, Moura SK. Comparison of frictional resistance between self-ligating and conventional brackets tied with elastomeric and metal ligature in orthodontic archwires. Dental Press J Orthod. 2014,19 (3): 114-119.
241. Monteiro MR, Silva LE, Elias CN, Vilella Ode V. Frictional resistance of self-ligating versus conventional brackets in different bracket-archwire-angle combinations. J Appl Oral Sci. 2014, 22 (3): 228-234.
243. Ozturk Ortan Y, Yurdakuloglu Arslan T, Aydemir B. A comparative in vitro study of frictional resistance between lingual brackets and stainless steel archwires. Eur J Orthod. 2012, 34 (1):119-125.
245. Pillai AR, Gangadharan A, Kumar S, Shah A. Comparison of the frictional resistance between archwire and different bracket system: An in vitro study. J Pharm Bioallied Sci. 2014, 6 (Suppl 1): S150-5.
246. Vijayalakshmi RD, Nagachandran KS, Kummi P, Jayakumar P. A comparative evaluation of metallurgical properties of stainless steel and TMA archwires with timolium and titanium niobium archwires–an in vitro study. Indian J Dent Res. 2009, 20 (4): 448-452.
247. Inami T, Tanimoto Y, Yamaguchi M, Shibata Y, Nishiyama N. Surface topography, hardness, and frictional properties of GFRP for esthetic orthodontic wires. J Biomed Mater Res B Appl Biomater. 2016, 104(1): 88-95.